The UAM’s Fenton Balloon is a unique component of the exhaust valve that provides an effective and safe way to control the breath cycles. In the rare case that the Fenton Balloon develops a hole or rip, or distorts in shape, it will not function properly and must be replaced.
These are the necessary items:
- A new Fenton Balloon
- A few drops of water
- Unscrew the tube assembly unit, which includes the top aluminum ring, the clear plastic housing and the valve assembly attached to the bottom.
- Turn using the top ring to make sure that it comes out as one piece.
- Make sure that the O-ring on the top does not drop.
- With fingers on the neck of the balloon, carefully peel off the balloon from its mounting.
- With a small amount of moisture on the neck, slide the new balloon on to its mounting and move it up so that the top ring fits in the groove on the mounting. â€¨
- Replace the tube assembly unit, taking care not to catch the balloon in the screw threads. Use a rotating action to make sure the balloon is fully visible inside the tube before locating the screw threads.
- Test the balloon by activating the bellows and observing the balloon inflate and deflate.
More Blog Posts
Product Note: COVID-19
We recognize the enormous pressure the COVID-19 pandemic places on health systems and healthcare providers as they work to...
2020 Year in Review: Crisis, Change, and Community
COVID-19 has demanded transformation and growth that could only have been achieved through a community’s commitment to deliver. Here are...
UAM Selected to Join Million Lives Club for Directly Serving More than 1 Million Patients
Gradian Health Systems is pleased to announce the Universal Anaesthesia Machine (UAM) has been selected as an official member...
Association of Anesthesiologists of Uganda and Gradian Health Critical Care Training
The Government of Uganda is undertaking an ambitious effort to increase critical care capacity by fully equipping ICUs at...