The UAM’s Fenton Balloon is a unique component of the exhaust valve that provides an effective and safe way to control the breath cycles. In the rare case that the Fenton Balloon develops a hole or rip, or distorts in shape, it will not function properly and must be replaced.
These are the necessary items:
- A new Fenton Balloon
- A few drops of water
- Unscrew the tube assembly unit, which includes the top aluminum ring, the clear plastic housing and the valve assembly attached to the bottom.
- Turn using the top ring to make sure that it comes out as one piece.
- Make sure that the O-ring on the top does not drop.
- With fingers on the neck of the balloon, carefully peel off the balloon from its mounting.
- With a small amount of moisture on the neck, slide the new balloon on to its mounting and move it up so that the top ring fits in the groove on the mounting. â€¨
- Replace the tube assembly unit, taking care not to catch the balloon in the screw threads. Use a rotating action to make sure the balloon is fully visible inside the tube before locating the screw threads.
- Test the balloon by activating the bellows and observing the balloon inflate and deflate.
More Blog Posts
Saving Lives at Birth: Project Microsite
Check out our new microsite for more information on our Saving Lives at Birth project! We’ll be updating it regularly as the...
Video: The UAM’s Impact at Kabwe General Hospital
At Kabwe General Hospital in Kabwe, Tanzania, the Universal Anaesthesia Machine (UAM) is regarded differently from other medical equipment....
Scaling Our Work in Tanzania with the Islamic Development Bank’s Transform Fund
When the Tanzanian government selected Gradian’s UAM as its anesthesia machine of choice for hundreds of health facility upgrades,...
What Our Clinical Trainers Can Teach Us
A week and a half ago, five of our clinical trainers came to visit our NYC offices. The team...