The UAM’s Fenton Balloon is a unique component of the exhaust valve that provides an effective and safe way to control the breath cycles. In the rare case that the Fenton Balloon develops a hole or rip, or distorts in shape, it will not function properly and must be replaced.
These are the necessary items:
- A new Fenton Balloon
- A few drops of water
- Unscrew the tube assembly unit, which includes the top aluminum ring, the clear plastic housing and the valve assembly attached to the bottom.
- Turn using the top ring to make sure that it comes out as one piece.
- Make sure that the O-ring on the top does not drop.
- With fingers on the neck of the balloon, carefully peel off the balloon from its mounting.
- With a small amount of moisture on the neck, slide the new balloon on to its mounting and move it up so that the top ring fits in the groove on the mounting. â€¨
- Replace the tube assembly unit, taking care not to catch the balloon in the screw threads. Use a rotating action to make sure the balloon is fully visible inside the tube before locating the screw threads.
- Test the balloon by activating the bellows and observing the balloon inflate and deflate.
More Blog Posts
CCV Service Note: Maintenance of CCV Battery
The CCV’s internal and external batteries must be maintained and status-checked regularly to ensure that maximum ventilation backup time...
Fast Company’s 2023 Innovation by Design Awards
New York, New York, August 22, 2023 – The K-O2 was honored in Fast Company’s Innovation by Design Awards...
Strengthening Healthcare in Tanzania: Gradian Health Systems Donates Three Intensive Care Ventilators (ICVs) to Muhimbili National Hospital
On July 11th, 2023, Gradian Health Systems donated three Intensive Care Ventilators (ICVs) to Muhimbili National Hospital in an...